Approx. Rs 15.85 Lakh / PieceGet Latest Price
|Minimum Order Quantity||1 Piece|
|Pressure Range||15 PSI|
|Mount Type||Floor Mounted Autoclave|
|Temperature Range||121 deg C|
|Automation Grade||Fully Automatic|
|Approval Certificate||ISO, EN 285, CE, FDA|
|Insulation Wall||Triple Wall|
|Country of Origin||Made in India|
Fully automatic PLC or Microprocessor controlled Horizontal Rectangular Autoclave (Steam Sterilizer), floor mounted with pre and post-vacuum treatment and with loading equipment. 200-250 ltrs capacity, operatable at 121°C at 15 PSI pressure
Door: The sterilizer supplied should be supplied with automatic sliding double door (electrically controlled) alternatively pneumatically controlled can be given, with door safety features.
Door Safety Systems:
Pressure sensor system should be available in the chamber to monitor the chamber pressure. Chamber should be completely depressurized before the door seal is retracted by vacuum
Door chamber should not be opened when chamber is pressurized.
A mechanical safety edge stops the door if it is obstructed while closing, thus protecting operator & loading equipment
A cycle should not start if the door is open or not properly locked.
The door seal should be made of silicon rubber gasket & on commencement of the process the door gasket is pressed against the rear force of the door by steam/air to ensure the door remains closed during the process.
The Sterilizer should be supplied with Automatic (Manual opening in case of automatic mechanism failure) vertical sliding door. Construction: Chamber & Doors: The chamber and doors should be made of solid, high quality 316L Stainless steel. The chamber should be jacketed to ensure the temperature uniformity in chamber with multiple heaters. Specify the number of heaters. The chamber floor is slightly sloped towards an internal drain to facilitate drainage. A stainless steel mesh strainer should be provided to protect the drain port from blockage by debris. The chamber is mounted on a stainless steel framework with height adjustable feet.Surface Treatment: Surface Treatment: The internal surface should be electro-chemically treated or mechanically treated for high quality smooth finish to facilitate cleaning. The resultant surface should be polished with 1.25 μm or less fineness to protect against corrosion. The internal corners should be rounded off to facilitate efficient cleaning. Insulation: The sterilizer jacket and door should be completely insulated to keep the autoclave cool on the outside. The insulation should be completely encased in a rigid removable sheet housing.Jacket The jacket should be made of 316L quality stainless steel with pressure gauge / digital display of jacket pressure Steam Generator: The sterilizer should have inbuilt steam generator of adequate capacity. In inbuilt steam generator should be made of 316L quality stainless steel The steam generator should have chloride free mineral wool/mineral glass wool of thickness 25 mm to 50 mm insulation with SS 316 or Aluminum. It should have a built in thermostat, pressure safety valve & water level glass gauge inspection device or water level indication on screen visible from service area. The heating element should be of sufficient capacity to make the sterilization process faster with maximum cycle time of 45-50mins in pre vacuum. It should also have the automatic blow down valve & degassing system for feeding water to steam generator. Pipes, Valves and Components: Pipes, Valves and Components: All the hot piping system should be made of Stainless Steel other pipes should be made of Stainless Steel / Brass / Copper All the process valves should be stainless steel or Copper Valves or Red Brass Valves & should be pneumatically/electrically operated piston valves for longer trouble free operations. All the non-standard components should be non-proprietary & should be easily sourced